World Of Controls understands the criticality of your requirement and works towards reducing the lead time as much as possible.
PWA135305-01 - Dual Vibration XY/Gap Monitor is available in stock which ships the same day.
PWA135305-01 - Dual Vibration XY/Gap Monitor comes in UNUSED as well as REBUILT condition.
To avail our best deals for PWA135305-01 - Dual Vibration XY/Gap Monitor, contact us and we will get back to you within 24 hours.
Part Number: PWA135305-01
Manufacturer: Bently Nevada
Product Type: Dual Vibration Monitor
Availability: In Stock
Country of Origin: USA
PWA135305-01 is a Dual Vibration XY/Gap Monitor manufactured by Bently Nevada and used in dual vibration monitor systems. The 3300/16 XY/Gap Dual Vibration Monitor accepts inputs from two proximity probe/Proximitor devices and continuously detects and monitors two independent channels of radial vibration and average shaft position (gap). The 3300/15 Dual Radial Vibration Monitor, which lacks radial position gap alerts, is intended to be replaced with this monitor. The real vibration has dropped below the setpoint level for the danger alarm. Contact your sales representative if the usual 4 Hz (240 CPM) low-frequency limit is insufficient for your application. For applications where the shaft rotative speed is less than 1,000 rpm, the 1 Hz (60 CPM) option is advised. Programmable by the user for +4 to +20 mA, 0 to -10 Vdc or +1 to +5 Vdc Voltage or current outputs are proportionate to the full-scale of the programmed monitor. There are separate recorder outputs per channel. Short circuits on recorder outputs have no impact on the monitor's operation.
Fig 1: Field Wiring Diagram
RADIAL VIBRATION: Radial vibration is characterized as the dynamic motion of the shaft that is perpendicular to the shaft centerline. Values for two channels are displayed by the Dual Vibration Monitor (Channel A and B).
PROBE GAP VOLTAGE: The probe gap voltage is the measurement of the gap between the face of a proximity probe and the surface being examined. It is a negative dc voltage. By activating the GAP switch, the probe gap value for each channel is displayed on the front panel meter.
OK - If the Proximitor output voltage falls within its upper and lower limits. The monitor relay drive to the system relay, as well as the channel OK LED, is both controlled by the OK detecting circuit.
OK RELAY - The Power Input Module contains the OK Relay. To activate the OK Relay, every channel in the rack needs to be OK or bypassed.
GAP ALARM: The GapAlert setpoints are displayed when the GAP and ALERT switches are pressed simultaneously. The ALERT LEDs turn on and the corresponding Alert Alarmrelay connections are actuated when the gap level is equal to or outside the over and under set point window limitations for 6 seconds.
World of Controls has the most comprehensive catalog of Bently Nevada Replacement parts. Our professionals are standing by to help you with your needs. If you require any additional information, please contact WOC immediately.
FREQUENTLY ASKED QUESTIONS
Does WOC sell to international Non-USA customers?
Yes, WOC offers services to clients around the globe. In fact, a large portion of our company is international, and we have experience shipping to locations all over the world.
Are the Products Genuine OEM Units?
These items have been certified by WOC Quality Control to be genuine OEM Units obtained through OEM or independent third-party channels.
Why Choose WOC Parts?
Industrial part sourcing is a specialty of the WOC team, We have years of experience locating uncommon and out-of-production equipment that our customers require in order to resume their operations. Although there are other businesses that make similar claims, we are certain that our dedication to quality and value is unmatched in our industry.